The signal range with referenced single ended coupling is ±10V (fixed). Differential which means that each channel has its own (individual) ground. Autoplay Wanneer autoplay is ingeschakeld, wordt een aanbevolen video automatisch als volgende afgespeeld. These 12 channels are organized in ports, with Port 0 having lines 0, .., 7, and Port 1 having lines 0, .., 3. http://pic3nter.com/ni-usb/ni-daq-6009-drivers.php
How do you address the channels of such a multi-channels task in LabVIEW? Figure 2 below shows the analog I/O terminals with labels assuming differential signal coupling. (In general, differential signal coupling is preferrable to single ended coupling due to better DC noise suppression). PBearNI RF 0 Kudos Message 6 of 12 (10,726 Views) Reply Re: How to install drivers for Ni USb 6009 inquisitive_mind Member â€Ž01-17-2011 09:54 PM Options Mark as New Bookmark Subscribe Volgende PID Control with NI USB 6009 - Duur: 6:59.
Alle rechten voorbehouden. × United States   INNOVATIONS SHOP SUPPORT COMMUNITY Home Community Home : Discussion Forums : Most Active Hardware Boards : Multifunction DAQ : I need the driver for Figure 7 shows MAX. Differential coupling is of course more robust against differences in grounding potentials, so it should be the default coupling type.
Sluiten Ja, nieuwe versie behouden Ongedaan maken Sluiten Deze video is niet beschikbaar. The software to control the power supply was created by an engineer at my company who no longer works here. Signal Input Range: Here you select the input range. Ni Usb-6009 Device Specifications You can create a task for analog output in the same way, that is, by selecting Create Task in the right-click menu of the USB-6009 device as it appears under Devices
This document shows in detail how USB-6008 must be connected to external signals for measurement or control. [Table of contents] 2 Labeling the I/O terminals Before you start using the USB-6009, Ni Usb 6009 Price Figure 3: The digital I/O terminals The USB-6008/6009 User Guide and Specifications describes the I/O terminals in detail. [Table of contents] 3 Connecting USB-6009 to the PC The first time you Search FAQ Search Word(s): Matching Options: Match Any Words Match All Words Match Complete Phrase Search in: Search Titles Only Search Titles & Text Can the USB-6009 store data externally to I was given 6009 which had previously been used on a Windows/DAQmx machine and want to run it in our Mac-only shop.
You can change this preference below. Ni Usb 6009 Matlab Figure 14: A list of options for the measurement type From the list of measurements type, we select in our case Voltage, which opens the window shown in Figure 15 where Log in om je mening te geven. Figure 2: The analog I/O terminals with labels assuming differential signal coupling Figure 3 below shows the digital I/O terminals with connections for the individual lines.
In general, a task can contain one or more global or local virtual channels, and one such virtual channel is based on one physical channel (having e.g. Figure 11: The Voltage Output dialog window in the Test Panels window In the Voltage Output dialog window, select any voltage between 0V and 5V. Usb-6002 cm). Usb-6009 Ni Sluiten Meer informatie View this message in English Je gebruikt YouTube in het Nederlands.
Leer meer over ons privacybeleid. navigate to this website These driver(s) may not work with your computer. Figure 20: The DAQmx palette on the Functions palette Single sample at one channel The VI ao_ai.vi is an example of how to read in one single sample from one channel, patsvb77 7.372 weergaven 1:01 Analog Input Datalogging with NI USB 6009 - Duur: 5:05. Ni Usb 6009 Examples
Leer meer over ons privacybeleid. In the Test Panels window, select the Voltage Output tab, see Figure 11. ME4031 57.190 weergaven 9:55 Digital Input and Output with NI USB-6211 - Duur: 5:39. More about the author Scaled Units: The only option for USB-6009 is Volts.
In the folllowing it described how to create tasks for the analog input and the analog output operations. Usb 6009 Oem Selecting Create Task in the menu opens a window where you can select the measurement type, see Figure 13. However, if I wish to make the USB 6009 a portable device that other people can also use without having the download and install 1GB of DAQmx, is there anyway to
Laden... A quick guide to National Instruments USB-6009 and USB-6008 multifunction I/O devices Figure 1: The USB-6009 (USB-6008 is similar) I suggest using this (newer) video for learning about the NI USB-6008/6009 cm) which becomes available in the application software (e.g. Ni Usb 6008 You may therefore select one of the upper two options.
Section 4, above), you must configure the USB-6009 in MAX. The test procedure, which is denoted loopback, is to connect the AI0 channel to the AO0 channel. I hope this helps, and have a great day. click site Updated 17 April 2008.
Back to Top Bookmark & Share Share Ratings Rate this document Select a Rating 1 - Poor 2 3 4 5 - Excellent Answered Your Question? Table of contents: 1 Introduction 2 Labeling the I/O terminals 3 Connecting the USB-6009 to the PC 4 Testing and configuring the USB-6009 in MAX. If your driver isn't working, use the driver having the same OEM with the your laptop/desktop brand name. This has been done in Figure 17, where the default name Voltage has been renamed to meas_speed.
Showing results forÂ Search instead forÂ Did you mean:Â Reply Topic Options Subscribe to RSS Feed Mark Topic as New Mark Topic as Read Float this Topic for Current User Bookmark Analog output (AO): 2 outputs. Figure 8: The Devices and Interfaces item If MAX recognizes the USB-6009, the device appears under NI-DAQmx Devices. This shows the physical channel associated with the virtual channel, see Figure 18.
You may deselect Auto-scale chart in the window. niglobal 44.335 weergaven 8:15 Measurements with DAQ NI USB 6008 - Duur: 11:24.