Subject: NIDAQ USB-6009 From: Rob Purser Rob Purser (view profile) 165 posts Date: 9 Sep, 2011 20:26:10 Message: 7 of 10 Reply to this message Add author to My Watch List This document explains how to test the device in Measurement and Automation Explorer (MAX). Voltage range is 0 - 5V (fixed). The most recent settings are saved, and are loaded the next time MAX is opened. check my blog
You may deselect Auto-scale chart in the window. Configuration of an analog input task See Figure 8. Leer meer over ons privacybeleid. Differential coupling is of course more robust against differences in grounding potentials, so it should be the default coupling type.
tasks, local and global virtual channels, physical channels, can be configured in MAX. Tags are public and visible to everyone. OK PRODUCTEN Bestelstatus en geschiedenis Bestel op artikelnummer Activeer een product Bestellen en betaling ONDERSTEUNING Dien een service request in Manuals Drivers Alliance Partners BEDRIJF Over National Instruments Over National Instruments Ask An Engineer Deze website maakt gebruik van cookies om uw bezoek aan de website beter te laten verlopen.
Figure 2: The analog I/O terminals with labels assuming differential signal coupling Figure 3 below shows the digital I/O terminals with connections for the individual lines. This document shows in detail how USB-6008 must be connected to external signals for measurement or control. [Table of contents] 2 Labeling the I/O terminals Before you start using the USB-6009, Tagging Messages can be tagged with a relevant label by any signed-in user. Ni Usb-6009 Matlab The price is 2100 NOK.
Alle rechten voorbehouden. Ni Usb 6009 Examples Below is a list of drivers that may be suitable for your device. The allowable signal ranges with differential coupling are ±20V, ±10V, ±5V, ±4V, ±2.5V, ±2V, ±1.25V, ±1V. Selecting Create Task in the menu opens a window where you can select the measurement type, see Figure 13.
Figure 5: The wizard has finished the installation of the driver software, and the USB-6009 is ready for use, after you click Finish (Norwegian: Fullfør) If you disconnect the USB-6009 from Ni Usb-6008 Driver Windows 7 Any help would be much appreciated. Input high is between 2.0V and +5.8V. cm) which becomes available in the application software (e.g.
We will perform a very simple test, which is sufficient if we are to check that both AO0 and AI0 work correctly. The Analog Input window should indicate the same (or almost the same) voltage as is set out on AO0. Ni Usb 6009 Datasheet Single samples at multiple channels In MAX you can define tasks which involves I/O on several (virtual) channels at a time. Ni Usb 6009 Price If you are looking for an update , pickup the latest one.
Top 3 Similar Drivers to National Instruments NI USB-6009 (138 Models) National Instruments NI DAQCard-6715 Multifunction DAQ | National Instruments National Instruments NI PCI-6013 Multifunction DAQ | National Instruments National Instruments click site All Rights Reserved. Thus, we will open MAX in the ordinary way via the Start menu on the PC. [Table of contents] 4 Testing and configuring the USB-6009 in MAX Here is a video Tasks and virtual channels can be given any name (e.g. Usb-6002
Can you see your devices in M&A Explorer? -Rob Subject: NIDAQ USB-6009 From: Garrison Garrison (view profile) 1 post Date: 9 Sep, 2011 18:03:27 Message: 6 of 10 Reply to this I typed daq.getVendors and got ans = Data acquisition vendor 'National Instruments': ID: 'ni' FullName: 'National Instruments' AdaptorVersion: '2.18 (R2011a)' DriverVersion: '9.3 NI-DAQmx' IsOperational: true I referred to the troubleshooting tips To actually perform the loopback test, right-click on the device labeled NI USB-6009: "Dev1" in MAX, and then select Test Panels, thereby opening the Test Panels. news The present document gives an example of how to use USB-6009 in LabVIEW.
Terminal Configuration: Referenced single ended which means that all channels have the same ground. Ni Usb-6009 Device Specifications Figure 20: The DAQmx palette on the Functions palette Single sample at one channel The VI ao_ai.vi is an example of how to read in one single sample from one channel, The DAQmx Write function writes the voltage value of the analog output channel as defined in task_ao.
For example, if AI1 is +10V and AI5 is –10V, then the measurement registered by USB-6008 is +20V. Thanks. 0 Kudos Message 1 of 4 (9,891 Views) Reply Re: I need the driver for NI USB 6009 so my computer will recognize the USB Brooks_C Active Participant â€Ž04-10-2008 01:01 You can also add an author to your watch list by going to a thread that the author has posted to and clicking on the "Add this author to my watch Usb 6009 Oem I'm sure that the 'nidaqmx.dll' file exists, and I've tried entering the entire filepath instead of just the filename.
With what you can learn with the DAQ device you will be able to more intelligently tweak/control your robot through conventional means for competition. Virtual channel list, which is the area having the Show Details button at the top. Newsgroup content is distributed by servers hosted by various organizations on the Internet. More about the author In Figure 11, an output voltage of 3.57V is set.