NI USB-6008 - Product Certification Dimensional Drawings NI USB-6008/6009 Accessory Kit - Dimensional Drawings USB-6008 - Dimensional Drawings Ask An Engineer Submit your question, view possible solutions, and post to the In these cases it is recommended that NI-DAQmx is used. NI-DAQmx Runtime 16.1.0 with Configuration Support - Windows 10, Windows 8, Windows 8.1, Window... Next, click the Analog Input tab in the Test Panels window, see Figure 12. http://pic3nter.com/ni-usb/ni-usb-6009-driver.php
The allowable signal ranges with differential coupling are ±20V, ±10V, ±5V, ±4V, ±2.5V, ±2V, ±1.25V, ±1V. Juridische informatie | © National Instruments Corporation. If a global channel is reconfigured, all tasks which involve that global channel are updated accordingly. You can use the Index Array function (on the Array palette).
Assuming that MAX recognizes the device, you can now run a self-test of it by selecting Self-Test in the menu. This has been done in Figure 17, where the default name Voltage has been renamed to meas_speed. patsvb77 6.344 weergaven 7:20 Programming Data Acquisition Applications with NI-DAQmx Functions - Duur: 6:20.
Mohon Bantuannya. - Discussion Forums mo beli ni usb-6008 - Discussion Forums mo beli ni usb-6008 - Discussion Forums Bagaimana Cara Pemodelan Motor DC dengan NI USB 6008 - Discussion Forums Scaled Units: The only option for USB-6009 is Volts. Sluiten Meer informatie View this message in English Je gebruikt YouTube in het Nederlands. Ni Usb 6009 Examples Max sampling rate is 48kS/s (48000 samples per second). 14 bits AD converter (USB-6008: 12 bits).
Figure 11: The Voltage Output dialog window in the Test Panels window In the Voltage Output dialog window, select any voltage between 0V and 5V. Ni Usb 6002 This document explains how to test the device in Measurement and Automation Explorer (MAX). task_motor and meas_speed, respectively), while the physical channels will have predefined names (e.g. cm).
The latter is available on the CD shipped with the device, and can alternatively be downloaded for free from http://ni.com (the NI-DAQ 8.0 file is approximately 0,5GB). Ni Usb-6009 Device Specifications Thus, we will open MAX in the ordinary way via the Start menu on the PC. [Table of contents] 4 Testing and configuring the USB-6009 in MAX Here is a video Laden... When the wizard has finished the installation of the driver software, the USB-6009 is ready for use (after you click Finish (Norwegian: Fullfør)), see Figure 5.
Sluiten Ja, nieuwe versie behouden Ongedaan maken Sluiten Deze video is niet beschikbaar. In the Test Panels window, select the Voltage Output tab, see Figure 11. Ni Usb-6008 Driver Windows 7 NI-DAQmx Runtime 17.0.0 - Windows 10, Windows 8, Windows 8.1, Windows 7, Windows Server 2012 R2... Ni Usb 6009 Price You may therefore select one of the upper two options.
NI recommends the USB-6001. 8 analog inputs (12-bit, 10 kS/s); 2 analog outputs (12-bit); 12 digital I/O; 32-bit counter DAQExpressâ„¢ interactive measurement software license Bus powered for high mobility; built-in signal navigate to this website Finn Haugen ([email protected]). Yes No Submit Deze website maakt gebruik van cookies om uw bezoek aan de website beter te laten verlopen. You can also, by right-clicking on the virtual channel name, define a (local) virtual channel to become a global virtual channel, which means that the channel can be a part of Usb 6009 Ni
NIDevZone 33.895 weergaven 5:39 Concrete station with NI usb 6008 and Labview - Duur: 1:32. My computer recognizes when the USB is plugged in although it asks me to install the driver for the USB. This tool will download and update the correct National Instruments NI USB-6009 driver versions automatically, protecting you against installing the wrong NI USB-6009 drivers. http://pic3nter.com/ni-usb/ni-daq-6009-drivers.php This driver software was installed along with the installation of the NI-DAQ software.
The DAQmx Start Task function must be invoked once, before the While loop starts. Ni Usb 6009 Matlab Stepper Motor Driver - Discussion Forums Stepper Motor Driver - Discussion Forums Training & Webcasts | VIEW ALL No content currently available Discussion Forum | VIEW ALL Bagaimana rangkaian sensor arus Laden...
All these items, i.e. If you select No Scale, it is the numeric value of raw measurement signal (V) that becomes available in the application software. aman0311tisf 2.290 weergaven 1:54 Introduction to National Instruments USB-6009 - Duur: 1:01. Usb 6009 Oem Digital input (DI) and digital output (DO): 12 channels which can be used as either DI or DO (configured individually).
If you had written a value outside this range to the DAQmx Write function you would get an error due to signal limitations function (it can only generate voltage outputs between This is done by selecting Add Channels in MAX, see Figure 17. The host system must also support the version of NI-DAQmx listed. click site Note how the task value (the magenta coloured line) is propagated through the relevant functions.
NI-DAQmx 17.1.0 - Windows 10, Windows 8.1, Windows 7, Windows Server 2012 R2, Windows Server 20... KnowledgeBase | VIEW ALL Can the NI USB-6008 OEM/ USB-6009 OEM be used without a USB Cable Connection? Toevoegen aan Wil je hier later nog een keer naar kijken? niglobal 44.335 weergaven 8:15 Measurements with DAQ NI USB 6008 - Duur: 11:24.
Custom Scaling: This gives you the possiblility to define a scaling of the raw measurement signal from the measurement unit (Volts in the USB-6009) to some other unit (e.g. USB CompactDAQ devices (except the cDAQ-9172) are supported on Pharlap RT systems with DAQmx 9.6 or later and LabVIEW RT 2012 or later. If you're using a version of LabVIEW newer than 7.1 then this is the version you want. Section 4, above), you must configure the USB-6009 in MAX.
The video replaces the present document. If you have an older version of LabVIEW then you'll need to check this KnowlegeBase article to find the correct version: NI-DAQ & LabVIEW Version Compatibility (for Windows) Once you Back to Top Bookmark & Share Share Ratings Rate this document Select a Rating 1 - Poor 2 3 4 5 - Excellent Answered Your Question?